GE Healthcare c Anesthesia Machine Technical Reference Manual Software Revision 1.x Approved −12−9 EET Junmin Ji (Rick) M c Product Data Sheet – GE Healthcare c Familiar, reliable, GE – with over years experience in anesthesia and quality leadership? 53 cm/21 in Depth: 40 cm/16 in Side of machine: cm/45 in Height: The c is the ideal solution for customers seeking an affordable, reliable and easy-to-use anaesthesia system. The c is designed and manufactured in.
|Published (Last):||25 May 2011|
|PDF File Size:||14.48 Mb|
|ePub File Size:||16.51 Mb|
|Price:||Free* [*Free Regsitration Required]|
After the last regulator has been tested, perform the checkout procedure Section 3. Remove the Software USB disk.
Follow the instructions on the screen to complete the installation. In mqchine section To ensure proper operation, the c Ventilator includes several tests that run automatically self tests and a series of menu pages that a qualified service person can use to test, calibrate, or troubleshoot ventilator related components in the c anesthesia machine Service Mode. Use care when moving or replacing system parts and components.
Set the system to On. Tighten the tester anestesia down onto the valve until the screw bottoms out on the top of the valve. Test the replacement switch assembly: The calibration status is displayed in the screen.
Examine the flapper and disk Cap for obstructions or debris. Control Sample Board under tabletop. GE Healthcare c Familiar, reliable, affordable Trust points? A pressure relief valve helps protect anesthedia system from high pressures. Mississauga, Ontario Australia Pty. Year of Establishment Exit the Service Mode and go to Normal Mode.
Front panel keyboard LED is part of it 2. Connect the tube from calibration pipe to the calibration fitting. Verify that T-handles are tight. Pull the knob E off the shaft of the encoder F. Complete steps 5 through 7 for each vaporizer amesthesia vaporizer position. Install the new disc seat into place. Install the valve block.
An electrical switch monitors the O2 supply pressure. Paw pressure gauge 7. Cylinder gaskets Use a leak detector or Snoop to check for source of leak. Remove the absorber, the vaporizers, gas cylinders, drawers and all auxiliary equipment.
Some internal parts have sharp edges and can cause cuts or abrasions. Remove the breathing system refer to Section 9.
Anesthesia Delivery – Products –
Connect an approved test device e. Disconnect the remaining cables at the four edges of the display Monitor Board monitor board.
It delivers the classic “Datex-Ohmeda clinical experience. Pull the knob and collar out from the front and remove the switch assembly. Remove the two M3 shoulder screws A that fasten the canister replacement latch lever C. Pull the tail C of the new free breathing valve flapper through the center of the valve sear until it locks in place.
The Breathing System Leak Test allows you to more precisely test the ventilator portion of the breathing system for leaks.